Today, Wuxi Jixiang Technology Co., Ltd. takes you to analyze why the data will jump frequently when using the infrared thermometer, mainly from the following points:
In this case, it is necessary to check whether the insulation of the connecting wire is damaged, and it is in short circuit to ground caused by regular contact with the shell of the machine.
The alignment of the linear infrared thermometer needs to be very good, but the parallelism can allow an error of ± 0.5mm, and the angle can allow an error of ± 12 °. However, if the parallelism error and the angle error are both too large, the display of digital jitter may occur. Then when this happens, the parallelism and angle must be adjusted.
The wiring of the infrared thermometer cannot be connected wrongly, and the power and output cables cannot be replaced. If the wires are connected incorrectly, there will be a large linear error. It is difficult to control, the control accuracy will be very poor, and the display is prone to jump.
Both FM interference and electrostatic interference may make the digital display of the linear infrared thermometer's digital beating. Therefore, the signal line of the electronic ruler should be separated from the strong current line of the equipment. The electronic scale must be grounded forcibly. The signal wire needs to be shielded, and a section of the electrical box should be grounded to the shield.
Wuxi Jixiang Technology Co., Ltd. is located on the shore of the beautiful Taihu Lake and the shore of Pohu Lake. It is a technology company specializing in infrared temperature measurement, infrared thermal imaging, and intelligent instrument technology. , Casting, forging, automotive parts, petrochemicals, building materials, rubber and plastics, new energy, electronics, automation, process monitoring, quality measurement, equipment diagnosis and maintenance and other major fields. Welcome to consult.
Related Industry Knowledge
- Application of Infrared Thermal Imager in Car Parts
- What is an infrared thermal imager
- Introduction of the scope of infrared thermometer
- Understanding and correct operation of infrared thermometer
- What are the advantages of using a non-contact infrared thermometer
- Principle and Application of Infrared Thermal Imager
- Analysis of infrared thermometer
- Operation process of non-contact infrared thermometer
- How the sniper escaped the detection of the thermal imager
- Can thermal imaging cameras really see people through walls?
- Update! Infrared Thermometer Purchase Elements Sharing
- Application of infrared thermal imager in electronics industry
- Brief introduction of infrared thermometer
- How to control the infrared thermometer to shut down unexpectedly
- What to pay attention to when using infrared thermometer
- Selection guide for very Xiang technology equipment maintenance thermal imager
- Five major reasons for affecting the accuracy of the red line thermometer
- Application of Infrared Camera in Building Energy Saving Field Detection
- Infrared thermal imaging temperature measurement system with belt transmission
- In the face of disaster, what role can infrared thermal imaging play?